Thanks everyone for the feedback and suggestions.
I wanted to give an update: The enlite sensor readings have stabilized though I'm still not sure they are as accurate as the sof-sensor was for my son but..... I just talked with a very knowledgeable medtronic rep (who wears an enlite) and I learned:
- by continuing to put the IV3000 over the sensor it is causing added pressure on the enlite (which is very sensitive) so that may be effecting his readings. (I did like the extra security 'blanket' of theIV 3000 but am willing to give the sensor-tape alone a try).
- the "Calibration Factor" is very important for the enlite (more than it was before). To calculate the CF: Meter Bg * 18/ISIG and it should be between 1.5 & 20 to calibrate but ideally between 5 & 9 for the most accurate readings.
Thanks again ....I haven't been able to talk TJ into the inner thighs but that sounds like a good place if we can get pass the Yikes factor!